The use of our devices at hospitals and medical practices is subject to stricter hygiene regulations than when used at home. Because the systems are used for different patients the following applies:
Adhering to a specific sequence of processing steps is particularly important in order to attain the high level of sterilisation safety required of sterile products.
You will attain the high level of sterilisation safety if you follow all the steps using efficient, validated procedures.
Cleaning and disinfection should be performed immediately after use. Put the completely disassembled nebuliser into an instrument washing machine on the 93 °C setting or place it in a cleaning and disinfecting solution.
After cleaning and disinfection, the nebuliser components must be put into a sterilisation pack that is appropriate for steam sterilisation and sterilised at 121 °C (up to 20 min.) or 134 °C (up to 7 min.).
The sterilised nebuliser must be stored in a dry, dust-free place away from sources of contamination.
Any questions? You can reach our service center at this number:
+44 (0) 1932 341122